A special form of the Pirani gauge is the pulsed Pirani vacuum gauge where the sensor wire is not operated at a constant temperature, but is cyclically heated up to a certain temperature threshold by an increasing voltage ramp. When the threshold is reached, the heating voltage is switched off and the sensor cools down again. The required heat-up time is used as a measure of pressure.
