  The MEG experiment at the Paul Scherrer Institut searches for the
charged-Lepton-Flavor-Violating mu+ -> e+ gamma decay. MEG has already set the
world best upper limit on the branching ratio: BR<4.2x10^-13 @ 90% C.l. An
upgrade (MEG II) of the whole detector has been approved to obtain a
substantial increase of sensitivity. Currently MEG II is completing the upgrade
of the various detectors, an engineering run and a pre-commissioning run were
carried out during 2018 and 2019. The new positron tracker is a unique volume,
ultra-light He based cylindrical drift chamber (CDCH), with high granularity: 9
layers of 192 square drift cells, ~6-9 mm wide, consist of ~12000 wires in a
full stereo configuration. To ensure the electrostatic stability of the drift
cells a new wiring strategy should be developed due to the high wire density
(12 wires/cm^2 ), the stringent precision requirements on the wire position and
uniformity of the wire mechanical tension (better than 0.5 g) The basic idea is
to create multiwire frames, by soldering a set of (16 or 32) wires on 40 um
thick custom wire-PCBs. Multiwire frames and PEEK spacers are overlapped
alternately along the radius, to set the proper cell width, in each of the
twelve sectors defined by the spokes of the rudder wheel shaped end-plates.
Despite to the conceptual simplicity of the assembling strategies, the building
of the multiwire frames, with the set requirements, imposes a use of an
automatic wiring system. The MEG II CDCH is the first cylindrical drift chamber
ever designed and built in a modular way and it will allow to track positrons,
with a momentum greater than 45 MeV/c, with high efficiency by using a very
small amount of material, 1.5x10^-3 X0 . We describe the CDCH design and
construction, the wiring phase at INFN-Lecce, the choice of the wires, their
mechanical properties, the assembly and sealing at INFN-Pisa and the
commissioning.
