  Micromanipulations, perfusions and measurements performed using glass
microelectrodes filled with an electrolyte is a conventional technique for
experimental morphological and membrane electrophysiological studies at a
single cell and membrane surface level. The typical (effective) diameter of the
end of the glass microelectrode is from 500 up to less than 100 nm, which
prevents one from observing it using a standard optical microscope in
accordance with the optical resolution criteria, since the diameter less than
500 nm is indistinguishable within the interference zone. Microprocessor
programming of the puller (microforge) that provides pulling and tearing allows
to obtain in certain regimes the adjusted diameter and shape of the
micropipette tip, although this result is not fully controlled due to the above
limitations. In this connection it is necessary to design the control devices
for the micropipette tips both at the preparation and operation stages
(intracellular or extracellular insertion). This method also should provide
visualization of the processes occurring upon interaction of the microelectrode
tip with the cell in real time, depending on the electrode type and state,
which allows to level the artifacts arising with the systematic error frequency
from the uncontrolled operation of the micropipette tip after different ways of
the microelectrode filling with the electrolyte. We propose an installation
scheme that solves the above problems by means of introducing an
interferometric device for microscopic control of the microelectrode and
micromanipulator or microperfusor, for the first time for a given type of
optical instruments combined with the interferometric optical scheme.
