  An all-fiber, micro-pulse and eye-safe high spectral resolution wind lidar
(HSRWL) at 1550nm is proposed and demonstrated by using a pair of upconversion
single-photon detectors and a fiber Fabry-Perot scanning interferometer
(FFP-SI). In order to improve the optical detection efficiency, both the
transmission spectrum and the reflection spectrum of the FFP-SI are used for
spectral analyses of the aerosol backscatter and the reference laser pulse. The
reference signal is tapped from the outgoing laser and served as a zero
velocity indicator. The Doppler shift is retrieved from a frequency response
function Q, which is defined as the ratio of difference of the transmitted
signal and the reflected signal to their sum. Taking advantages of high
signal-to-noise ratio of the detectors and high spectral resolution of the
FFP-SI, the Q spectra of the aerosol backscatter are reconstructed along the
line-of-sight (LOS) of the telescope. By applying a least squares fit procedure
to the measured Q spectra, the center frequencies and the bandwidths are
obtained simultaneously. And then the Doppler shifts are determined relative to
the center frequency of the reference signal. To eliminate the influence of
temperature fluctuations on the FFP-SI, the FFP-SI is cased in a chamber with
temperature stability of 0.001 during the measurement. Continuous LOS wind
observations are carried out on two days at Hefei (31.843 N, 117.265 E), China.
In the meantime, LOS wind measurements from the HSRWL show good agreement with
the results from an ultrasonic wind sensor (Vaisala windcap WMT52). Due to the
computational expensive of the convolution operation of the Q function, an
empirical method is adopted to evaluate the quality of the measurements. The
standard deviation of the wind speed is 0.76 m/s at the 1.8 km. The standard
deviation of the retrieved bandwidth variation is 2.07 MHz at the 1.8 km.
