  Signal to noise ratios (SNR) in magnetic resonance microscopy images are
limited by acquisition times and the decreasing number of spins in smaller
voxels. Significant SNR gains from cooling of the RF receiver are only realized
when the Johnson noise generated within the RF hardware is large compared to
the electromagnetic noise produced by the sample. Cryogenic cooling of imaging
probes is common in high field systems but proves difficult to insulate the
sample from extreme temperatures. We designed a chamber to cool only the tune
and match circuitry to show it is possible to achieve much of the available SNR
gain available for cooled coils. We designed a microcoil circuit to resonate at
650 MHz for imaging on a 15.2 T scanner. Surface loops and solenoids of varying
diameters were tested to determine the largest diameter coil that demonstrated
significant SNR gains from cooling. A liquid N2 cryochamber was designed to
cool the circuitry while leaving the RF coil in ambient air. As the cryochamber
was filled with liquid N2, Q-factors were measured on the bench while
monitoring the coil's surface temperature. Improvements of SNR on images of
solutions were demonstrated via cooling the tune and match circuit in the
magnet bore. At 650 MHz, loops and solenoids < 3 mm in diameter showed
significant improvements in quality factor on the bench. The resistance of the
variable capacitors and the coaxial cable were measured to be 45% and 32% of
room temperature values near the Larmor frequency. Images obtained with a 2
turn, 3 mm diameter loop with the matching circuit at room temperature and then
cooled with liquid nitrogen demonstrated SNR improvements of a factor of 2. By
cooling the tune and match circuit and leaving the surface loop in ambient air,
SNR was improved by a factor of 2. The results are significant because it
allows for more space to insulate the sample from extreme temperatures.
