  Since large dose are administrated to the patient in brachytherapy treatment,
ensuring the right dose is delivered is highly critical. This study presents a
first step in solving the standing issue of accurately knowing the dosimeter
position at all time during in vivo dosimetry. In this work, an energy
independent dosimeter, namely plastic scintillation detector, is coupled to an
electromagnetic (EM) sensor having sub-mm positional accuracy for real-time
tracking of the dosimeter position. However, adding an EM sensor adds materials
in the path to the scintillator and thus could potentially perturb the dose
measurements.To confirm the perturbation presence, 4 different sensors were
placed in front of the scintillator so the radiation does not arrive to it
directly. Variation of the distance between the sensor and the scintillator was
used to quantify the effect on the signal at 0{\deg} and 90{\deg}.To test the
dependence for each sensor, the signal measurement were taken from 0{\deg} to
90{\deg} with 10${\deg}$ increment. The 5PCBDOF sensor showed an increased
signal of almost $20{\%}$ with increasing beam angle. Sensor 5DOF, 5DOFthin and
6DOFshowed no significant angle dependance.The 6DOF and 5DOFthin sensor's cable
revealed no extra signal attenuation. The latter gives a smaller overall
attenuation. Therefore, the 5DOFthin is chosen to be part of the novel
dosimeter construction. It has a jitter error of $\pm$ 0.06 mm and a
reproductibility of$ {\pm}$ 0.008 mm. In the optimal operating range, the
average positional uncertainty is less than 0.2 mm. Average angle errors are at
most of 1.1${\deg}$. It is feasible to integrate an EM tracking sensor to an
energy independent plastic scintillation dosimeter with minimal impact to the
collected signal as well as sufficient positional accuracy to keep dose
uncertainty below 5${\%}$.
