  The objective was to study uncertainty in antenna input impedance resulting
from full one-port Vector Network Analyzer (VNA) measurements. The VNA process
equation in the reflection coefficient p of a load, its measurement m and three
errors Es -determinable from three standard loads and their measurements- was
considered. Differentials were selected to represent measurement inaccuracies
and load uncertainties (Differential Errors). The differential operator was
applied on the process equation and the total differential error dp for any
unknown load (Device Under Test DUT) was expressed in terms of dEs and dm,
without any simplification. Consequently, the differential error of input
impedance Z -or any other physical quantity differentiably dependent on p- is
expressible. Furthermore, to express precisely a comparison relation between
complex differential errors, the geometric Differential Error Region and its
Differential Error Intervals were defined. Practical results are presented for
an indoor UHF ground-plane antenna in contrast with a common 50 Ohm DC resistor
inside an aluminum box. These two built, unshielded and shielded, DUTs were
tested against frequency under different system configurations and measurement
considerations. Intermediate results for Es and dEs characterize the
measurement system itself. A number of calculations and illustrations
demonstrate the application of the method.
