quality:correction, quality:corruption, information:classification, linguistic:translation
Magnetic spectrometer

When a fast charged particle (charge q, mass m) enters a constant magnetic field B at right angles, it is deflected into a circular path of radius r, due to the Lorentz force. The momentum p of the particle is then given by

where m and v are mass and velocity of the particle. The focusing principle of the oldest and simplest magnetic spectrometer, the semicircular spectrometer, invented by J. K. Danisz, is shown on the left. A constant magnetic field is perpendicular to the page. Charged particles of momentum p that pass the slit are deflected into circular paths of radius r = p/qB. It turns out that they all hit the horizontal line at nearly the same place, the focus; here a particle counter should be placed. Varying B, this makes possible to measure the energy spectrum of alpha particles in an alpha particle spectrometer, of beta particles in a beta particle spectrometer, of particles (e.g., fast ions) in a particle spectrometer, or to measure the relative content of the various masses in a mass spectrometer.

Since Danysz' time, many types of magnetic spectrometers more complicated than the semicircular type have been devised.