What is the minimum and maximum range of measurement in micro-sieverts per hour?
883620
It goes from 0 - 999 microsieverts. The radiation I took pegged the machine out at the 999. From how I used the machine, it seemed pretty accurate. And yes, it will bounce around at the low end as .18 microsieverts is pretty low to begin with. Love the machine and will use it again if I am in need of another Radioactive Iodine treatment. I feel comfortable knowing that the readings in the house are now normal.
883621
Spec sheet claims that range of measure of indicated background radiation level is 0.03 to 1,000 mcSv/h, although if it measures above 1.2 you should seek shelter.
883622
The parameters are adjustable.l think mine came set at 1.2 max at the moment.There is also a graph showing a broader spectrum.Readings here are normally from .07 to .14,although l have detected some pasta products with slightly higher readings.There is a clicker which can be turned off,a reliability bar,color screen etc.A very nice instrument
2280012
As with all DMMs 0 ohms is the lowest. Accuracy is 0.08% (per specs). The range dial shows 200 ohms as the lowest selector setting for measurement 0 to 200 ohms.
2195831
The minimum on/off would be one minute. The max would be 23 hours 59 minutes.
1928079
A time-domain reflectometer (TDR) will do it. Probably very expensive though. Go to this Wikipedia link for images and how (what principle) the instrument uses in layman terms. https://en.wikipedia.org/w/index.php?title=Time-domain_reflectometer&oldid;=584192273
554544
I would guess, it being made of metal, that it would not be long. maybe 5 minutes, tops. my units read 2 degrees F less than the more accurate digital clocks. How much variance are you getting?
10139
A minute is the smallest marked increment of time. You could probably do less, but it would loose accuracy.
688131
I measured a strip with my mm ruler and it would be very close to 3mm, only a slight bit more if any!
1998178
Sorry, this one won't be suitable for such accuracy. For making effective low-ohms measurements you're going to really want something that does 4-wire Ohms measurements as the IR losses in the measurement wires will be significant if the total resistance under test is so low. Unfortunately those do tend to run several hundred dollars at a minimum.
1998177
Sorry, this one won't be suitable for such accuracy. For making effective low-ohms measurements you're going to really want something that does 4-wire Ohms measurements as the I·R losses in the measurement wires will be significant if the total resistance under test is so low. Unfortunately those do tend to run several hundred dollars at a minimum.
32516
No. The meter is set for microwave-oven frequency over a certain range. I think it is too microwave oven-specific for accurate cell phone radiation readings.
746086
The meter has two range settings: Low which measures 30 to 100 dB and High which measures from 60 to 130 dB
223628
The answer is provided by the manual at http://www.bosch-professional.com/gb/en/gms-120-23317-ocs-p/ which specifies: Digital Detector GMS 120 Article number 3 601 K81 0.. Maximum scanning depth* - Ferrous metals - 120mm - Non-ferrous metals (copper) - 80mm - Live conductors 110-230 V - 50mm (voltage applied)** - Wood - 38mm Regards, Mikhail
1628093
I bought those too, the readings with the no-calibration strips still came out as 0.1's almost all the time with occasional 0.2's just like the first set of calibrated strips that I had bought with the meter originally. I'm an engineer and don't know exactly how they do it, but nanoscale manufacturing these days is doing amazing things so maybe the calibration sample is on the strip sub-microscopically and we simply can't see it. Just follow the instructions and be happy.
32514
Have no idea. All we know, it wasn't accurate for the micro-wave oven. We returned it
84896
7 - 100, according to the product description. I have only used mine for metric measurements.
1662735
I've had values ​​up to 1.35 A with on good sunny days. Normal values ​​of 0.8 to 1.1.It's easy to measure current Isc, as you point out, with the multimeter in 10A range, and placing the test leads to the cable ends of the panel, + and -. It is desirable that the multimeter has a resolution of 10 mA.
1874308
My best guess is everything under 200hz gets rolled off. Not sure on the high range.